The OP5705-IO simulator offers quick, single-ended connections, through RJ45 and mini BNC connectors, to any monitoring device (i.e. oscilloscope, etc.). These mini BNC jacks let you monitor 4 channels individually. Simply follow these instructions (as illustrated in the figure below):
The monitoring board has a gain of 0.1 to allow a broader range of signals to be read. The oscilloscope must be adjusted with a gain of 10 to compensate and ensure accurate readings.
Only connect Ethernet cable with RJ45 ends from the upper section (A) monitoring jacks to the lower section monitoring panel (B, as shown). Connecting any other cable or device may result in damage to the equipment.